M80 ultrasound head
1. Applied gel on fleshy parts (eg.abdomen,hip,thigh) The amount depends on
of the ultrasound head.
2. Set work time (10 min every part), Adjust output intensity (10weak~50 strong).
3. Click “Start”, Beautician use M80 working head to start working.
4. Beautician hold the ultrasound head to move slowly on skin, in circle or beeline
repeat,the other hand push the fat towards the ultrasound head.
5. Do not use on back, avoid using on bones.
6. Do not use on uterus part when women do abdomen care . Abdomen treatment
Menstruate should be avoided.
* Enough gel are needed, you may feel painful if medium is not enough.
* The total treatment time for daily care should be within 30 min.
7. The machine will pause automatically if the set ting time is over.
8. Clean the leftovers with hot towel use warm towel to clean the ultrasound head.
RF Function procedure
1.Choose RF function ,set work time(each part 20 minutes),
regulate output intensity:1(weak)~10(strong).according to parts need
cared. Both Tripolar and Bipolar RF head can use for face ,arms and body,bipolar RF
head for eye and nose wing.
2.Applied gel on skin need treated,click start/pause ,the machine will start working.
3.Move the electrode on skin slowly,do not stay unmoved.
4.The machine will pause automatically if the setting time is over.
5.Clean the leftovers with hot towel, use warm towel to clean the RF electrode head.
6. Use warm water to clean skin.
7.If you want to change to other functions within the setting time during the operation,
press pause and then press  to Change, readjust work time, output intensity and
8 . Sterilize the used heads.
1.When start RF function,make sure that the lines not tangled.
2.Enough gel is needed to avoid pains.
3.Do not wear any aglet, both for the beautician and the users. do not use on parts
with metal infilling inside.
4.Do not make the RF electrode stay on any part without Moving.
Voltage: AC110V/220V, 50Hz /60Hz
Input Power: ＜110W
Output power of Cavitation: 50W
Ultrasonic frequency: 40KHZ
RF frequency :5MHZ
GW: 15KG NW:10 KG
2.IC card 2pcs
3.User manual 1pc
4.Ultrasonic cavitation slimming head pc
5.Bipolar RF head 1pc
6. Tripolar RF head 1 pc
7.Power line 1pc
8.Accessories shelf 2pcs
9.Hexagonal screws 8pcs
10.Hexagonal screwdriver 1pc